Often on the shop floor you would like to verify the settings of the gage you are using.There are several simple methods to ensure your settings stay correct. Today’s tip covers the MRP® crest diameter and ovality gage and coupling inspection. When using an MRP® crest diameter gage to inspect crest diameter on a coupling, […]
Would you like to receive the Gage Report in your email? Sign up below.
For Tubing and Casing, properly setting a TH-3002R thread height gage is easy, yet often done incorrectly. The setting standard (1014) has a square-shaped notch and a v-shaped notch. First, use the square-shaped notch, zero the gage. Then, rotate the standard and use the v-shaped notch to verify the gage is correctly zeroed. If the […]